Re: [IPk] Non delivery
I am relatively new to pumping and have never heard about lubricating
reservoirs.? Should I be doing this?? I use the Medtronic 522 pump.?
I did get delivery problems with Quicksets every time I had them in my abdomen,
it worked during?priming when disconnected,?so I knew it was the site. Strangely
enough Quicksets were?ok in my leg.? I have since started using Silhouettes
and?have not had a single problem.? They feel better as well.??
From: Diana Maynard <email @ redacted>
To: email @ redacted
Sent: Thu, 31 Jul 2008 12:59 pm
Subject: Re: [IPk] Non delivery
Ray, are you using QuickSets? These have notorious problems with no delivery
alarms (though some people do get on fine with them).?
First thing to check is whether it's a site issue or a pump/reservoir issue.?
Prime when disconnected and see if you still get the alarm, to check this.?
If it's a site issue, and you're using QuickSets, try something else (eg
Silhouettes). It's unlikely to be a pump issue but it is possible. Also check
your reservoirs are in date and you're lubricating them properly first.?
Su Chadda wrote:?
> I have had this a couple of times. The first time I had to return my > pump.
The last time it happened - I changed the set and took the > batteries out?
> and it was ok.?
> Pumper Since 2005?
> ----- Original Message ----- From: "Ray Jordan" <email @ redacted>?
> To: <email @ redacted>?
> Sent: Thursday, July 31, 2008 12:02 PM?
> Subject: [IPk] Non delivery?
>> Hi all, has anyone had trouble with their pump regarding non delivery if?
>> so what did you do to solve the problem ?.?
>> I have been experiencing this problem just lately; it is a nuisance?
>> especially when it occurs during the night.?
>> Ray Jordan pumper since Jan 2004?
for HELP or to subscribe/unsubscribe/change list versions,