[Previous Months][Date Index][Thread Index][Join - Register][Login]
[Message Prev][Message Next][Thread Prev][Thread Next]
Re: [IP] Re: Minimed A-35 alarm
- Subject: Re: [IP] Re: Minimed A-35 alarm
- From: email @ redacted
- Date: Thu, 22 Oct 1998 17:44:10 EDT
In a message dated 98-10-21 16:54:54 EDT, you write:
What does the A-35 alarm indicate? I always thought that was the display
you get after a pump "reset".
A-35 alarm is a delivery safety check. most often occurs when the lead screw
is turn when the pump is in the run mode. MiniMed specifically states to call
them if it occurs more than once without explaination.
Insulin-Pumpers website http://www.bizsystems.com/Diabetes/