[Previous Months][Date Index][Thread Index][Join - Register][Login]
[Message Prev][Message Next][Thread Prev][Thread Next]
Re: [IP] resouvir question
OK All - time to share the foolproof de-bubblization (sorry for the
new word) technique.....here we go (and I am talking MM terms
1. make sure the insulin is room temp
2. hold the reservoir/syringe vertically - move the plunger up and
down a few times to spread the lubricant - then pull it back to the
desired amount you are going to fill.
3. hold the insulin vial vertical - and insert the needle straight up
into the insulin.
NOW - THE KEY!!!
Inject the air into the insulin. Put resistance against the plunger,
and DO NOT PULL IT - let the air pressure move it back against
the resistance of your thumb. Let it fill about 50 units - then push it
all back in. Let the air pressure push out about 50-75 units again,
then push it back in. this time, give it a couple of flicks on the side
to loosen any bubbles - and push it back in again (tightly to the
top). Now, allow the air pressure to push it back (again against
resistance) until the pressure won't push any further. then, and
only then - do you pull the plunger out slowly to the desired
Try it and tell me how it works. We NEVER (I shouldn't say never -
RARELY) have bubble problems.....
> I have a question though, as much as I love my pump, i'm not looking
> foward to the set change tomorrow. I've done two already. Both times,
> it took forever. I had a really hard time getting all the air bubbles
> out of the syringe.
Steve Winer - Dad to Allie
14 y/o - dx 6/97 - MM508 12/99
for HELP or to subscribe/unsubscribe, contact: HELP@insulin-pumpers.org
send a DONATION http://www.Insulin-Pumpers.org/donate.shtml