[Previous Months][Date Index][Thread Index][Join - Register][Login]
[Message Prev][Message Next][Thread Prev][Thread Next]
Re: [IP] Animas Occlusion Alarms
well I don't know why they did it but I have an explanation.
The 'occlusion' is trying to tell you that something is blocking the
pumping mechanism and that no insulin is delivered.
I am sure that you would like to know what is blocking it and how it
can be resolved. So the first thing that I would want to do anyway is
take a look at my tubing and get the pump going manually to see the
insulin squirts coming out of the tubing. Well, and to get the insulin
going means that you would either give yourself a bolus and use the
I guess Animas choose "priming" for multiple reasons but mainly because
it resets the pump in its default operating state (cancels the temp
basal, for example). Assuming the worst - which corresponds to the most
conservative assumption that you are fixing things anyway I think it is
better to be safe than sorry.
Of course, after having said all that, I usually just prime and if it
looks fine, meaning insulin is coming out very easily, I just reconnect
to continue what I was doing before. I think I get more alarms if I use
the shorter tubing. But you might want to check your occlusion limits
in the setup section. I have set mine to 'High' meaning that the pump
is more sensitive to 'occlusions' or pressure increases.
Hope it helps,
> Can anyone explain to me why my R1000 insists on thinking that >it
> needs to
> reprimed when I get an occlusion alarm?
> I got woken up twice last night with occlusion alarms (12am and >2am).
> I was sleeping on top of the infusion site and the pump didn't >appear
> the additional pressure I was placing on the site. I just
> quick-connect on my comfort and reprimed and reattached after >each
> There was no way I was going to change the site out at 2am since >I had
> put it in earlier in the day. (I was tired, I drove almost 400 >miles
> This morning, the site appears happy and bg levels are normal. >I'm
> what the thought process the people who designed the pump >software
> determining that it's necessary to reprime after an occlusion >alarm.
> think a good explanation for this behavior?
for HELP or to subscribe/unsubscribe, contact: