[Previous Months][Date Index][Thread Index][Join - Register][Login]
[Message Prev][Message Next][Thread Prev][Thread Next]
On Sun, 4 Jan 1998, Mike Maturen wrote:
> I am a Disetronic user. The last three infusion sets (Tender) have
> caused occlusion alarms. The cartridge and tubing are fine. When I
> remove the cannula from the site, it is clogged with a whitish material.
> It appears waxy, but crumbles when handled.
> In 6 years of pumping, I have never seen this. Neither has the
> Disetronic service group. It has happened on the last three sets.
> Any ideas?
> Mike Maturen