[Previous Months][Date Index][Thread Index][Join - Register][Login]
[Message Prev][Message Next][Thread Prev][Thread Next]
[IP] strange occlusion in cannula
I am a Disetronic user. The last three infusion sets (Tender) have
caused occlusion alarms. The cartridge and tubing are fine. When I
remove the cannula from the site, it is clogged with a whitish material.
It appears waxy, but crumbles when handled.
In 6 years of pumping, I have never seen this. Neither has the
Disetronic service group. It has happened on the last three sets.