[Previous Months][Date Index][Thread Index][Join - Register][Login]
[Message Prev][Message Next][Thread Prev][Thread Next]
Re: [IP] Minimed 506 no delivery alarm
The problem may be related to wear on the plunger drive mechanism, or
even just lint and dust collecting in the grooves. This may make it
more difficult for the plunger to depress, creating an alarm. Carefully
inspect and clean the "lead screw" which depresses the plunger and see
if this resolves the issue. You may want to also perform a lead screw
rotation test. If this does not resolve the issue, contact minimed.
Janine Shea wrote:
> Hey gang,
> I have recently been getting a no delivery alarm when I get down to 40
> units left in the syringe. It has happened the last 4 times, at least. I
> turn it off and the pump goes back to giving me insulin. At first I
> changed everything, then I decided to see what happened when it kept
> occurring. Everything seems fine, blood sugars consistent.
> This has never happened in 5 years I have had this model. No delivery
> alarms have only occurred when I was right out or for a couple of
> Any thoughts? I am finding it convenient, because it reminds me I am
> getting low, but sort of wondering if I should talk to Minimed.
> Vancouver, BC, Canada
> email @ redacted
> Insulin-Pumpers website http://www.bizsystems.com/Diabetes/
Insulin-Pumpers website http://www.bizsystems.com/Diabetes/